Search the Community
Showing results for tags 'frequency measuring continuous'.
Found 1 result
Hi Dear Digilent Team, A question concerning the measurement capabilities ot the Analog Discovery 2 and WaveForm software: Is it possible to measure continously and "without losses" a frequency of an analog or digital entry? Goal is to log a frequency over a long period of time between EACH following flanks Optionnaly also measure times between flanks of different channels Examples: - Measure the freq/time between each following rising flanks of a channel - Measure the freq/time between each following falling flanks of a channel - Measure the freq/time between each following flanks (rising and falling) of a channel ... Most scopes measure frequency only "around" the trigger flank Thanks for each feedback Regards