I recently went to use my Analog Discovery 1 and I found that it was giving me all sorts of strange results that I did not expect. I figured that the calibration must have drifted since I hadn't used the device in a year or so, but when I performed a calibration it was still significantly off. Here's a link to an album showing comparisons between the AD1's readings versus a multimeter: https://drive.google.com/open?id=0B2q6FpLmYUaVb1dQSFpIQVJYWXc . I don't really know what to do at this point, but perhaps someone here has run into this issue before and can recommend a fix. Thanks!
Edit: For what it is worth, I have tried using different USB cables, different ports on my computer, different computers/operating systems, different jumper wires, and a different breadboard all to the same effect.